VEINOPLUS ELECTRODES PACK
Self-adhesive and reusable
Veinoplus® PACK electrodes are accessories intended to be used with all Veinoplus electrical stimulators. The pack contains one pair of biocompatible self-adhesive and reusable electrodes (size 8×13 cm (3 »x5 »), OVAL shape).
We are often asked about the on/off performance of the electrodes. The answer is very simple: The durability of our electrodes depends entirely on the application, storage, humidity and care on the part of the user.
Application: Before each electrode application, skin must be cleaned with water and soap to remove any lotions, skin oils, make-up and dead skin. Any debris on the skin will be transferred to the gel of the electrode thereby compromising the integrity of the gel’s adhesiveness. Therefore, a clean surface is the most important factor in the number of applications for an electrode.
Storage: After every use the electrode must be returned to the « ON » side of the Mylar liner, and placed in the original, re-sealable package to prevent loss of moisture. The electrodes should be stored at a temperature that would be comfortable for a human being, or somewhere between +5°C/41.0°F and 27°C/80.6° F. with humidity at 35-50%.
DO NOT : Store in freezer/refrigerator, leave in extreme heat, submerge in water, or leave outside of the package.
Care: After usage, if the electrode picks up small amounts of debris you may use a drop of water on your finger to gently rub the debris off the electrode. Most important to the care of the electrode is applying it to a clean surface with each use.
DO NOT : Use soap, alcohol, submerge in water, or scrub with abrasive material.
The number of applications depends entirely on the proper application, storage and care by the user. The durability of the electrodes depends on the adhesiveness of the gel and its adhesiveness depends on keeping the gel clean, hydrated and free from foreign debris.
Yes. When electrodes show wear and tear, or are no longer comfortable for the patient, simply replace them with a new set.
For proper hygiene, the FDA requires them to be labeled for single patient use. Our electrodes are intended for single patient use only. By sharing electrodes, you are subjected to cross contamination. All electrodes are disposable and should be discarded if compromised, and a fresh set used.
The products are designed for reuse provided there is proper care of the skin and the electrodes. However, with reuse, the risk of contamination increases with each application. Each time an electrode is removed, it lifts off dirt and skin debris, which in turn becomes absorbed into the gel. To maintain optimal hygiene you would want to change electrodes when they become contaminated in this fashion. The pH of the gel is controlled at a level to prevent the spontaneous growth of molds. However, during use in a spore rich environment it is possible to overwhelm this mold growth control feature of the gel.
The definition of hypoallergenic is « less likely to cause an allergic reaction than other comparable goods. » There is no scientific formula to measure or test this; therefore the gel is not labeled as hypoallergenic.
Electrodes are latex free. No latex of any kind is used in the manufactured electrodes or in the packaging.
Always lift electrodes from the edge, NOT the leadwire.
Use of hot or cold packs for long periods of time can cause adhesive separation.
The electrodes are made only in the USA.
SDS – Hydrogel for Veinoplus® PACK electrodes